View Catalogue AP-V80 Series Durable Multi-Fluid Digital Pressure Sensors Catalogue. Two Digital Display Pressure Sensor Amplifier; AP-V80W Series. A rugged and compact sensor head has been assembled using a single-piece stainless steel pressure element and housing. The AP-V80W Series can be used. Cheap sensor mouse, Buy Quality sensor clock directly from China sensor engine Suppliers: KEYENCE pressure sensor AP-V80W Enjoy ✓Free Shipping.
|Published (Last):||27 September 2009|
|PDF File Size:||7.6 Mb|
|ePub File Size:||3.12 Mb|
|Price:||Free* [*Free Regsitration Required]|
AP-V80W Series Pressure Sensor Applications : AP-V80 series | KEYENCE America
Amplifier unit NPN type: The lower part indicates the bottom value. Press together the body and the connector cover into which the cables were inserted to make a connection. It turns on if the value external input is turned on exceeds the setting value B2. Pressure difference while the zero-shift is turned on. It turns off if the external input is turned on hysteresis margin is subtracted from the setting value B1.
To dismount the sensor, raise the main body in the direction of the arrow 3 while pushing the main body in the direction of the arrow 1. Check the load and adjust it to the rated range. Displays the highest the lowest peak bottom value of pressure while the peak bottom hold is working.
KEYENCE pressure sensor AP-V80W
The setting value indicator 2 illuminates when the setting value is P2. Residual voltage 1 V max. The lower part v80e the setting values P1, P2, B1 and B2. It is valid only when the zero-shift is turned on.
Durable Multi-Fluid Digital Pressure Sensors
If no operations are made for 3 minutes in this state, indicators return to the Eco display again. In high-resolution mode, the value is indicated in kPa.
Current value display and buttons. Turns off when the pressure is out of the range. F-1 General-purpose mode This is a mode where the pressure detection point can be specified at desired two points. Zero-shift This is a function that sets the pressure at that time forcibly to zero by inputting an external signal.
F-3 Leakage test specific mode This is a mode suitable for a leakage test. Output 1 Detects the difference between the pressure value of the sensor inserted in the H higher pressure value side and the one inserted in the L lower pressure value side.
Setting value indicator 2 illuminates when the setting value is P2. The pressure difference during the zero-shift input is displayed as zero in the current setting value while the zero-shift input is turned on.
It turns on when the pressure value goes lower than the setting value P1. The display changes to the unit change screen when the button is released. The button pressed at first on the Eco display will become invalid. Does not initialize the setting value.
P2 Hys B2 Hys P1 Hys B1 Hys To the display mode Note If you change the already selected sensor head to a different type, be sure to perform initial setting before setting b80w sensor head type to use. The setting value indicator H-L flashes when the setting value is B1. Cut the cable to an appropriate length About About 16 mm and strip away about 16 mm of the 16 mm outermost insulation.
In high-resolution mode, the value is indicated in mbar. Keeps displaying the peak bottom value since the MODE button was pressed. Note that switching of the display remains valid. Leakage test 20 20 Zero-shift input Zero-shift input — 0. Use a pair of pliers or similar tool to press them together. Do not use the product in an environment with flammable gas or something similar. The output indicator and the alarm indicator function normally.
V80d value that has undergone ZERO reset is stored even if the power is turned off. F-2 Double window mode AP14S K This mode is used to judge whether the pressure value is within the range between the upper limit and lower limit values at the detection point.
When [ErrH] is displayed sensor head is unconnected or disconnected. The detection points can be switched by turning the external input on or off. S or more is applied while readjust sensitivity. The hysteresis margin is changeable in all modes setting.